The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||9 August 2012|
|PDF File Size:||20.56 Mb|
|ePub File Size:||5.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
Component replacement and internal adjustments must be made only byqualified service personnel.
Refer to the Safety Summary page at the beginning of this guide for general safetyinformation. OCP The overcurrent protection state is enabled.
Page 26 3 – Installationat the load. You can follow any responses to this entry through the RSS 2.
Keysight Technologies (Agilent HP) B used or new for sale at used-line
The next problem was the fact that the front panel meter was a little off from what I was measuring at the output. On the Function keypad, press Output. If you want to quickly view the content of pages jp on the following pages of the manual, you can use them.
Disabling it will result in some disabled or missing features. The voltage drops across theload wires should be limited to less than two volts. Selftest error messages appear as: Ho can leave a responseor trackback from your own site.
Agilent / HP 6632B 100 Watt DC System Power Supply 20v 5a
I have some old scopes gp fans in them, and the b is typically quieter than most of them. Amazon Renewed Refurbished products with a warranty.
The output of the dcsource may be adjusted to any value within the boundaries shown. Although my original post about this is pretty old I still use this same unit as my main bench supply. Connect the appropriate current monitor as shown in figure B-1A.
Press Shift, 6632g,Enter8.
Front Panel Operation – 52 – Querying and Clearing Output ProtectionThe dc source will disable its output if it detects an overvoltage or overcurrent fault condition. Set the instrument to the operating state that you want to save. Other 662b fault conditions such as overtemperature also will disable the output. C – Error Messages—Invalid suffix [unrecognized units, or units not appropriate]—Suffix not allowed—Invalid character data [bad character, or unrecognized]—Character data too long—Character data not allowed—String data error—Invalid string data [e.
A – SpecificationsTable A Figure B-1 shows the test setup. Page 44 5 – Front Panel OperationAs previously mentioned, the HP A dc source has the capability of measuring outputwaveform parameters such as peak, minimum, high level, and low level as illustrated in thefollowing figure.
Connect the DMM in dc mode across the current shunt. Learn more about Amazon Prime. Record this value andthe current reading on the front panel display.
Set the output current to its full scale value see Table B After some head scratching and a LANG change this was resolved. Refer to Table to calculate the voltage dropfor hhp commonly used AWG copper wire. Withoutabox Submit to Film Festivals. Page 75 Manual UpdatesThe following updates have been made to this manual since the November printingindicated on the Printing History page.
Failure to comply with these precautions or with specific warnings elsewhere in this manual violates safetystandards of design, manufacture, and intended use of the instrument. You can change the password when the dc source is incalibration mode which requires you to enter the existing password.
Using the Keypad shift pandSome of the front panel keys perform two functions, one labeled in black and theother in blue. Page 56 B – Verification and Calibration5. The dc source must 662b turn-on selftest before you can proceed with theverification tests. After much swearing, compressed air, and windex, the inside of the chassis looked like this:.
Turn-On Checkout – 4Procedure6.