Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||16 December 2010|
|PDF File Size:||9.34 Mb|
|ePub File Size:||11.62 Mb|
|Price:||Free* [*Free Regsitration Required]|
To determine the actual surface temperature of a device, both the heat-sinking and steady-state errors must be considered. Milton Freewater, Oregon, United States. Use care not to place excessive strain on these wires.
You are commenting using your WordPress. Do this by selecting the AC mA function and 2 range. Image not available Photos not available for this variation. It is recommended that the contents of this section be read and understood before attempting to operate the instrument. The A display will indicate the battery voltage; between 4.
Seller assumes all responsibility for this listing.
Fluke A Handheld “Bench” Multimeter Repair/Restore – Lazy Electrons
Since a DMM has no earth ground connection in itself, the earth ground connection must be suplied from the voltage source. Disconnect the input from the V-H input. The mathematical formula for determining the rms value of an ac voltage is: If the specified equipment 0840a not available, other equipment having equivalent specifications may be used.
The A display should read between tluke. Check the waveforms at these points as follows. This fuse is located behind the mA input jack and is removed by pressing in lightly on the jack then turning it counterclockwise! Select the AC V function and 2 range.
The oscilloscope display should be similar to waveform A4. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab.
Estimated on or before Wed. This item will ship to United Statesbut the seller has not specified shipping options.
Accumulation of dust or dirt particles between the output terminals of the Model 80F-5 can be removed using clean dry pressurized air. General Maximum Common Mode Voltage: The oscilloscope display should be similar to waveform A3. This item will be shipped through the Global Shipping Program and includes international tracking.
Showing Results for Fluke 8040A
Install the Model 80F front cover and strain relief. Please reference the instrument model number when requesting a new shipping container. Contaminates can be washed from the circuit board with demineralized water and a soft brush avoid getting excessive amounts of water on the switches. This item will be shipped through the Global Shipping Program and includes international tracking.
For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab. Here it is, twist and take it out. Federal Supply Code for Manufacturers. Physical design of the Model 80F allows direct mating to the input terminals of the Fluke voltmeters. These solutions will react with the plastic materials of the instrument.
Reverse flexing of these connectors can damage the solder joints. Unplug the battery power wires from the A circuit board. I liked the form factor and the bench meter appeal, but still it could be handheld. Select the KCl function.
John Fluke Model 8040A Digital Multimeter
Select the AC V function then refer to Table for the ranges and inputs for the performance check. Box Beaman Place Zip: In the event the part you dluke has been replaced by a new or improved part, the replacement will be accompanied by an explanatory note and installation instructions, if necessary.
Other offers may also be available.
The battery can be normal cell or rechargeable Ni-Cd. Any individual digit lfuke illuminated. Access Information The oscilloscope display shall be similar to waveform B2.
To avoid damaging the MOS devices you should 1 minimize handling, 2 discharge personal static before handling the device, 3 handle the MOS device by the body; avoid touching the pins on the device, 4 do not slide the MOS device over any surface, 5 avoid touching plastic, vinyl or styrofoam when working with MOS devices, and 6 always use a grounded soldering iron when installing or removing MOS devices. Box N. Here are the internals, very interesting design, and beautiful construction.
Select the ranges and apply the inputs indicated in Table Tip the bottom edge of the front panel up to clear the switches.